3 Wire Thread Measuring System Applications:
- Certify set plug gages and working thread plug gages
- Monitor the wear on working thread plug gages
- Monitor and control pitch diameter variation during thread fabrication
- Use in conjunction with “GO and “NOT GO ring gages to control thread size to the most demanding specifications
- Determine out of roundness and taper that may exist in threaded parts
- Applications for preplating and postplating thread measurement
- Eliminate the cost and time involved in using outside calibration services
- Reduce measurement time to a fraction of the time normally taken using the traditional three-wire method
Thread measuring wire holders,micrometer stands and micrometers avaiable.
THREAD MEASURING WIRES
These wires are the best sizes for standard 60° inch and metric threads. They may be used with other thread forms or with special pitches, constants for which may be calculated from the formula shown below. They are
made from selected high speed steel, are extremely accurate in roundness and straightness and are within.000020″ (20 millionths) of exact size.
The formula for determining wire constants for all 60° threads = 3W – .866025P. Where W = wire size and P = Pitch (1/T.P.I.) The most accurate and universally recognized method obtaining pitch diameter measurements of thread plug or setting plug gages is by means of using three hardened steel measuring wires laid into opposite sides of the thread, as shown in the diagram. Precise measurements with proper pressures over these wires minus the appropriate constant equals the measured pitch diameter.